It includes a “Function Panel”. This is where I get error D 00 40 where after rading through the pages of the manual I got to the conclusion that D stands for data error, first two characters, xx, stand for X axis, the second set of 2 characters Y axis Did you try other commands which take a longer time to execute? It recognises the equipment by setting it to the default GPIB address Transportability across different computer platforms 2. There are several advantages to using the dll form of the driver.
|Date Added:||24 October 2017|
|File Size:||31.34 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
You must also provide an include path for hpb.
Agilent 4156 Analyzer
Message 5 of 6. After detailing a bit more the process do you still think there might be a connection drop out between the Labvjew interface and HP?
Two elements of the standard do not apply to the instrument since it is not a VXI instrument. I’ll 4516b an example of sampling code I use later today.
Most Active Software Boards: I will at least try that, I tried working with two other 4156v as well and somehow sampling is way harder than sweeping. I have done many kinds of sweeping on this one, but this is my first time with sampling and I can’t seem to figure that one out.
Message 6 of 6. If i were you, i would contact KeySight formerly known as Agilent formerly known as HP what that message on the device is meaning. Message 1 of I really appreciate all the feedback I can receive. Did you get this working? Message 10 of Do not build the driver in the directory in which it was installed.
Agilent Analyzer – IEEE (GPIB) Driver for LabVIEW – National Instruments
Do you have any experience with this? I have no experience with LAbView kabview far. Message 9 of It’s alright, I now have a code that works perfectly! After this I can see the cursor on the bottom of the I V graph on the monitor of the sliding from left to right when settinp a Sweep of Voltage of SMU1 from 0 to 1 V, pts.
Agilent Technologies / Keysight Technologies 4156B
This is where I get error D 00 40 where after rading through the pages of the manual I got to the conclusion that D stands for data error, first two characters, xx, stand for X axis, the second set of labvjew characters Y axis The data I want to represent is IV curves.
I will try nevertheless to find a solution.
Sampling on HP I have a very old VI with a very old driver. I look forward to your feedback! A list of contact information for other countries is available on the Agilent Technologies website at http: I would like to know if there is any VI available for 2 terminal devices in this case labvjew wafer diodes so that I can control directly the llabview from LabView and be able to read out the measured data directly from the HP via LabView?
You can right click on the error cluster which shows error 6 and select “Explain Error”. I am not sure if the Instrument Driver for the B adds custom messages. Moreover I cannot see the trace of the sweep I expected a trace of labviww MA constant current and voltage ranging from 0 to 1 V not on the monitor of neither on the graph plot of the LabView VI interface.
Norbert Norbert CEO: Moreover on the third printscreen with the graphics plot it can be seen that there is an error from HP B code: At the same time I get an error on the monitor of D00 A small update on the setup definition I use: